Compact, economical, and easy-to-use, the SAVe II+ is built for dependable use during a crisis. Medical personnel have relied on AutoMedx products devices for over a decade. The new SAVe II+ delivers up to 50% more flow than the SAVe II and can generate up to 20 cmH2O of PEEP. The SAVe II+ replaces the SAVe II.
This kit includes the following:
- 1x SAVe II+ Ventilator
- 1x Hard Case (water and dust proof), PN F20065
- 1x Power Charger, PN M42090
- 1x Patient Circuit, PN F20066 (3 year shelf life)
- 1x Extendable Oxygen Reservor Tubing, PN F20072
- 1x Noise Attenuator, PN M41112
SAVe II+ Product & Performance Specifications
Dimensions Ventilator 6.5" x 6.25" x 2.0" (81 cubic inches)
Weight Ventilator 2.8 lbs. (1.3 kg)
Kit 9.9 lbs. (4.5 kg)
Operating Mode A/C-CMV
Mode Description Time cycled, pressure limited, volume targeted
PRESET Heights 4' 3" - 6' 3" in 3 inch increments
PRESET Target 6 ml/kg of IBW or 30 cmH2O of Working Pressure
Flow Rate 0 - 40 LPM
I:E Ratio 1:2 Fixed
Respiratory Rate 8 - 30 BPM
Tidal Volume 200 - 800 ml
Peak Inspiratory Pressure 10 - 60 cmH2O
PEEP 0 - 20 cmH2O
Inadvertent PEEP < 2.0 cmH2O
FiO2 Range: 21 - 100%
Oxygen Input: 0 - 12 LPM from Flow Regulated Source
Inspiratory Trigger 2.0 cmH2O
Inspiratory Resistance < 6 cmH2O @ 30 LPM
Expiratory Resistance < 6 cmH2O @ 30 LPM
Dead Space < 100 ml
Battery Duration 8.25-9.25 Hours (see below note)
Power Input 100 – 240 VAC / 50-60 Hz
Audible Alarms Meets 60601-1-8
Shock & Vibration IEC 60068-2-6 and 60068-2-36
Ingress Protection Vent IP24
Protective Case IP67
Breath Rate (BPM) 8 – 30
Tidal Volume (ml) 200 - 800
Pressure Limit (cmH2O) 10 - 60
PEEP (cmH2O) 0 - 20
PIP (cmH2O) 0 - 60
PEEP (cmH2O) 0 - 25
I:E Ratio 1:2
Inspiratory Time (Ti) 0.67 - 2.50 seconds
Expiratory Time (Te) 1.3 - 5.0 seconds
Flow Rate (Avg LPM) 4 - 40
Control Unit 8 years
Breathing Circuit 3 year shelf life (single use)
1. All specifications include a tolerance of ±10% of nominal value unless stated otherwise in the Operator’s
Manual. Test conditions available upon request.
2. Delivered Tidal Volume may be materially affected by very low lung compliance (< 0.01 L/cmH2O)
3. With a fully charged new battery the 5’9” preset will run approximately 8.25 hrs with (2600 mAh battery) or 9.25 hrs with 2800 mAh battery)
Limited Warranty Applicable to the SAVe II+
AutoMedx warrants to the original purchaser (“Customer”) of the SAVe II+ that if there is a defect in material or workmanship in the SAVe II+ and AutoMedx is notified of such defect within three (3) years of Customer’s original purchase, AutoMedx shall, in its sole and absolute discretion, repair or provide a replacement of such defective part(s) at no charge to the Customer, provided that this warranty provision is not applicable to batteries or used consumables.
Limited Warranty Applicable to the Battery
The life of the battery is materially affected by many factors. As such, AutoMedx warrants to the Customer of the SAVe II+ that, if there is a defect in material or workmanship in the battery contained in the SAVe II+ and AutoMedx is notified of such defect within one (1) year of Customer’s original purchase, AutoMedx shall, in its sole and absolute discretion, repair or provide a replacement of such defective battery at no charge to the Customer.
The sole remedy for a defect in materials or workmanship of the SAVe II+ (or the battery or any other component of the SAVe II+) shall be, at AutoMedx’s sole and exclusive discretion, repair or replacement of the defective SAVe II+ or component thereof, as the case may be.
AutoMedx’s warranty shall not apply to defects or conditions resulting from: (a) repairs by an unauthorized party; (b) improper maintenance; (c) modifications made without written permission of AutoMedx; (d) damage by accident, abuse, misuse, or misapplication; or (e) operation otherwise than in accordance with this manual or other instructions furnished by AutoMedx.
AutoMedx’s warranty shall not apply if the unit has been disassembled.
AutoMedx’s warranty shall not apply to: (a) any Product if the serial number of such Product has been altered, defaced or removed or (b) any used consumables.
AutoMedx’s warranty is neither assignable nor transferable. All warranty repairs shall be subject to return postage billing.
Disclaimer of Warranty and Limitation on Remedies
THE WARRANTY AND REMEDIES SET FORTH ABOVE ARE EXCLUSIVE AND IN LIEU OF ALL OTHERS, WHETHER ORAL OR WRITTEN, EXPRESS OR IMPLIED. AUTOMEDX SPECIFICALLY DISCLAIMS ANY AND ALL IMPLIED WARRANTIES, INCLUDING WITHOUT LIMITATION, WARRANTIES OF MERCHANTABILITY AND FITNESS FOR A PARTICULAR PURPOSE.
AUTOMEDX IS NOT RESPONSIBLE FOR DIRECT, INDIRECT, SPECIAL, INCIDENTAL, CONSEQUENTIAL OR EXEMPLARY DAMAGES WHETHER BASED ON CONTRACT, TORT, OR ANY OTHER LEGAL THEORY.
To the maximum extent permitted by applicable law, in no event shall AutoMedx or its Suppliers be liable for any special, incidental, indirect, physical or consequential damages whatsoever arising out of the use or inability to use the SAVe II+ product and or accessories. In any case, AutoMedx’s entire liability shall be limited to the amount actually paid for the purchase of the SAVe II+ product. Valid proof of purchase required.
Some countries, states, or provinces do not allow the exclusion or limitation of implied warranties or the limitation of incidental or consequential damages for certain products supplied to consumers, or the limitation of liability for personal injury, so the above limitations and exclusions may be limited in their application to you. When the implied warranties are not allowed to be excluded in their entirety, they will be limited to the duration of the applicable written warranty. This warranty gives you specific legal rights, which may vary depending on local law.